View and Download Gossen MetraWatt PROFISIX instruction manual online. hand-held exposure meter. PROFISIX Measuring Instruments pdf manual. You have an older device from GOSSEN? Here you can find the corresponding operating instructions for previous devices in the field of PROFISIX. Condition: Good The guide is in a good condition, edgewear, minor tear, and scratches on the cover. The inside leaves.
|Published (Last):||3 November 2015|
|PDF File Size:||5.51 Mb|
|ePub File Size:||17.12 Mb|
|Price:||Free* [*Free Regsitration Required]|
Gossen Profisix -or- new light meter.
I have alwasy assumed that my Profisix worked the way others did mine is serial 3D Amazon Restaurants Food delivery from local restaurants. JDMvWApr 24, The computer dial incorporates standard exposure compensation, exposure factor compensation for filters or close up work, as well as I – IX steps for zone exposure. Null pointer ambient and flash meter Display: If however there is a difference between the gosesn, this difference will be seen as a current running through the spool, generating a magnetic field that profisox the needle accordingly.
No, create an account now. Would you like to tell us about a lower price? The Profisix, known as the Luna Pro SBC in some countries, is an excellent meter with reflected and incident capabilities. This scenario is not very likely.
Gossen MetraWatt PROFISIX Instruction Manual
Welcome to apug Jmqx. Example of complete set. I like and don’t like the size. Sorry Jasper but you are quite wrong the numbers either side of the 0 used for the null reading are whole stops, and clearly marked as such in the meter’s handbook.
Gossen PROFISIX instruction manual, user manual, PDF manual, free manuals
I’ve only had mine for bossen years but it has worked flawlessly all the time. This by the way is linear and that is why the distance between the numbers isn’t but this is outside the discussion.
The problem with this approach is that if you do not have the necessary precision in the first place all you do is multiply the error without gaining control.
It is eminently repairable, although in many cases replacement with another used one may be more cost-effective. Yes, my password is: Jesper, I don’t think you may have explained it quite correctly, maybe you are misunderstanding the Profi-six meter and the dial s on the unit.
But they fit your hand glssen gracefully, one-handed operation is a treat. Aug 22, 2. Yes, my presbyopic eyes like my Luna Pro. Together with the adapters the Profisix forms a complete light measurement system. The dial has three separate rings:. I will not turn this into “I am right, you are wrong” discussion there are other forums for that.
PREVIOUS DEVICES | Gossen
Whatever it is, I must say that I prefer the added precision this has given. Some 10 different adapters are available for a broad range of applications like the Tele manuao, reproduction adapter, LAB adapter for darkroom appl.
Adjust the ratio and you have adjusted the movement. Flash sync speed Aperture range Exposure time Cine.
Set the ASA speed Step 2. East Dane Designer Men’s Fashion. Personal tools Log in.
As I said, I have no obvious reason to believe that my meter is faulty unless by sheer coincidence the fault only applies to the movement of the needle and that all readings are still correct. The Profisix is an excellent meter. Amazon Rapids Fun stories for kids on the go. Somehow I think you either have a different or doctored scale, or your meter is a bit different to any other Profi-six meter I have seen. Aug 23, 8. The needle is in the middle null position.